Keithley Instruments, Inc. warrants this product to be free from defects in material . aspects of Nanovoltmeter operation, refer to the Model User’s Manual. A. Nanovoltmeter. Accessories Supplied. Low Thermal Input Cable with spade lugs, m (4 ft). User manual, service manual, contact cleaner. View and Download Keithley user manual online. Nanovoltmeter. Measuring Instruments pdf manual download. Also for: a.
|Published (Last):||16 April 2016|
|PDF File Size:||7.6 Mb|
|ePub File Size:||8.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
PRESet enables continuous initiation.
Table Input cable color codes Analog Output Accuracy Be sure that DC voltage measurement accuracy is within required limits before attempting this procedure.
Parts lists The electrical parts lists for the Model are shown in tables at the end of this section.
Providing the finest test equipment solutions since 1992
Table of Contents Add to my manuals Add. Troubleshooting Introduction This section of the manual will assist you in troubleshooting and repairing the Model Triggering Device action The primary device action is a measurement.
Voltage and Temperature Measurements Perform the following steps to enable or disable line cycle synchronization: Interface Function Codes Figure Measurement Event Register Aborting Calibration The calibration procedures described keeithley are generally the only instrument calibration required in the field.
A programmable delay is available after event detection. Ratio Delta — Explains how to perform Delta measurements, which are used to cancel the effects of thermal EMFs in the test leads.
Instrument Reassembly Disassembly Instrument reassembly Reassemble the instrument by reversing the previous disassembly procedures. Connections The Model Input Cable, which is a supplied accessory, is terminated with a LEMO con- nector on one end and copper lugs on the other end. Doesn’t that kinda raise a flag? Purpose To request the calibration constants for the active range.
Power transformer removal Perform the following steps to remove the power transformer: Comments to this Manuals Your Name.
KEITHLEY USER MANUAL Pdf Download.
Ratio and Delta For even greater precision, the Relative feature of the Model can be used to null out thermal EMFs, which can corrupt low voltage measurements. Service personnel are trained to work on live circuits, perform safe installations, and repair products.
Table DC voltage measurement accuracy limits However, the following procedure is provided for those who wish to independently measure temperature measurement accuracy. On the Modelpress the REL key to zero the display. Disassembly Disassembly Case cover removal Follow the steps below to remove the case cover to gain access to internal parts.
Software requirements In order to use the calibration programs, you will need the following computer software: Carefully pry the locking tab that keihhley the fuse carrier to the power module. The sample counter is decremented to 4 causing operation to loop back to Device Action for a measurement on Channel 1.
Cleaning Test Circuit Connectors Make sure kiethley the copper contact surfaces are free of oxidation before making the connection. Got it, continue to print.
Keithley A Dual-Channel Ultra-Low Voltage Nanovoltmeter- at the Test Equipment Depot
See Section 5 of this manual for details on basic Delta measure- ments. Storing Readings In Buffer Internal Stepping Stepping and Scanning Operation: Model front panel was pressed. Manua Pc Boards