Absolute and relative reliability of isokinetic and isometric trunk strength testing using the IsoMed dynamometer. Roth R(1), Donath L(2). extension measurements using the IsoMed system, Isokinet Exerc Sci knee extension: A reproducibility study using the IsoMed dynamometer. The purpose of this study was to determine relative as well as absolute reproducibility for the IsoMed dynamometer in measuring Peak.
|Published (Last):||20 October 2013|
|PDF File Size:||13.44 Mb|
|ePub File Size:||8.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Abstract The purpose of this study was to determine relative as well as absolute reproducibility for the IsoMed dynamometer in measuring Peak Torque PT during concentric and eccentric maximum knee extension at two angular velocities.
Click on the images to get more details on the right.
Concentric and eccentric isokinetic knee extension: Based on the systematic errors found for flexor measurements and a clear improvement in reproducibility parameters iso,ed T1-T2 to T2-T3 for nearly all measurements, we recommend the use of a familiarisation session prior to actual testing.
Several investigations have reported reproducibility for isokinetic single-joint SJ knee strength measurements. Other measuring chambers e.
No further or recurring license fees are due! Systematic assessment of muscle strength of the lower extremities throughout the annual training cycle in athletes is crucial from a performance perspective for the optimization of the training process, as well as a health perspective with regard to injury prevention.
Filter by Years All 0. The response is checked by repeatedly measuring the test source entered in the system which is used for calibration and comparing ismed mean value with the actual target value.
The results of all measurements can be represented in tabular form. Reproducibility of concentric and eccentric isokinetic multi-joint leg extension measurements using the IsoMed system.
Abstract Systematic assessment of muscle strength of the lower extremities throughout the annual training cycle in athletes is crucial from a performance perspective for the optimization of the training process, as well as a health perspective isomsd regard to injury prevention.
The measuring chamber is connected via USB interface.
A high Isomeed was evident in the eccentric mode in both groups highlighting a need for particular strengthening. Abstract This study investigated the effect of a week recreational skiing intervention on lower limb muscle strength and cardiorespiratory fitness in participants with unilateral total knee arthroplasty TKA. Nuclide and activity of the test source, however, are freely selectable.
Reproducibility calculations revealed ICC-values of 0. Our PC-based Dose Calibrator ISOMED is designed for the fast and accurate determination of the activity or volume activity of radiopharmaceuticals that 20000 used in the nuclear medicine for diagnostics and therapy.
The intervention group skied three times a week whereas the control did not change their daily routines over a period of 3 months.
In conclusion, isomec reproducibility was shown for the IsoMed dynamometer in measuring PT during concentric and eccentric knee extension. No effect was observed for cardiorespiratory fitness. Thirty four physically active male subjects mean age: The ISOMED protocols meet the requirements of the medical authorities to provide evidence for the quality checks carried out. Isokinetics and Exercise Science.
A reproducibility study using the IsoMed dynamometer. Isokinetic devices are highly rated in strength-related performance diagnosis. With regard to low strength ratios a prevention programme should be suggested for both observed groups of professional female handball players to reduce the risk of injury.
In case of using a TFT monitor with touch-screen-function, you can work completely without mouse and keyboard. A sample of 33 healthy young subjects age: These adaptations remained unchanged toward the retention test.
In addition to the above mentioned complete systems, we also offer a basic module consisting of measuring chamber and ISOMED program. We use a holder for that, mounted between the monitor and the original monitor holder.
However, to date, only a few studies have concentrated on multi-joint MJ leg strength measurements. If you need any further information on our RPM software, please call us or use our contact form! Repeated measures analysis of variance ANOVA was applied for detection of systematic errors over sessions. Moderate to iosmed reproducibility was shown for the IsoMed system in measuring PF during concentric and eccentric isokinetic leg extension.
Detection of systematic errors was performed using paired sample t-tests, relative and absolute reliability were examined using intraclass correlation coefficient ICC 2.
Therefore, we hypothesized that this kind of intervention mitigates the loss of muscle mass and strength associated with ageing and disuse and reduces the side-to-side asymmetry in older TKA patients.
Above all, the long-term quality and user-friendliness of the predecessor models served as a basis for the current model. In order for an isokinetic device to be clinically useful, the reliability of specific applications isomdd be established.