Download FLUKE PM PM PM PM 2X2MV,MHZ OSCILLOSOPE SM service manual & repair info for electronics experts. Re: Fluke PM manual needed (MHz scope) The service manual for Fluke / Philips PM PM PM PM can be found. PM ). Deflection Manual positioning: Horizontal, Vertical, Both . Web access: com.
|Published (Last):||14 January 2004|
|PDF File Size:||17.97 Mb|
|ePub File Size:||5.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
This diagram shows the Y-output amplifier. Otherwise the component is located on the side with small components. No lot shall be removed without the authority of the Auctioneer and unless under the supervision of the Auctioneer’s servants or agents.
There are really no sign of flames anywhere.
Oscilloscope Fluke PM powersupply problem?
Before purchasing or ordering replacement parts, check the parts list for value, tolerance, rating and description. Every attenuator basically consists of four sections.
Dangerous Substances It is expressly brought to the attention of Buyers and potential buyers that certain types of plant or main service installations could contain blue and white asbestos, dangerous chemicals and hazardous waste which if not handled correctly during their removal from the premises could be in breach of the Health and Safety at Work ActSections Control of Substances Hazardous to Health Regulations COSHH or other current legislation regulating the use of manuall substances in a working flukee.
If the fuse was intact, it’s unlikely. All instruments have four vertical channels. Refer to section 8.
The driver transistors are V and V The Consumer Protection Act No lots are sold as new. This is present on all 4 channels.
Oscilloscope Fluke PM3082 powersupply problem?
Input Characteristics Note 1: Please also note that some lots are located on the first floor so please ensure that sufficient resources are in place for the removal of your lots. Handle the CRT and its side connections carefully.
Please note time changed to end from Turn on the oscilloscope under test. The horizontal MTB deflection coefficients can be varied in steps such as done in 6.
Then take care that its screen is pressed tight to the front side of the chassis.
Main Time Base Description. Page The horizontal MTB deflection coefficients can be varied in steps such as done in 6. Accept Cookies Learn More. However, this begs the question of what caused the flames and smoke? Control of the illumination amplifier V takes place by means of the signal DAC2, level about 1. Pardon my suspicious nature, but methinks you should run it with the covers off and keep an eye on this scope for a fluie. The linearity of the vertical amplifier is checked by moving a signal with a fixed amplitude vertically over the entire screen area.
Probe Adjust signal is distortion-free and has equal intensity. Come back when bidding opens to start bidding. When installing the CRT, first remove its protective cover.
The highest bid at the close of bidding subject to approval by the Auctioneers and Vendor also marks a conclusion of a contract of sale between the Vendor and the Buyer. Biasing current for these outputs is provided via V, V and R Page Verify that the deviation from the ideal straight line does not exceed 0. Sorry, bidding has ended on this item. You are the current highest bidder! Where satisfactory evidence is not provided, goods will be subject to UK VAT at the standard rate of export and fkuke deposit will not be refunded.
The x1 is switched by control signal PA1X1.
Risk and Title 7. Page The oscilloscope must be capable of displaying signal amplitudes that are larger than the screen. The amplifier is located at the small printed circuit board at the CRT socket. If not present install 2 penlight LR6 back up batteries in the holder at the rear panel of the oscilloscope. Manua, of options Y-out and external trigger See also section 9.
Fluke PM Analog Oscilloscope | eBay
If there is any inconsistency between any of these conditions and anything in the Notices to Purchasers, then those in the Notices to Purchasers shall prevail. Some of these voltages remain some time after disconnecting the instrument from the mains. This eliminates calibration inaccuracies due to temperature changes.