Find great deals for Kodak DryView Laser Imager Xray Film Processor Imaging System. Shop with confidence on eBay!. Find great deals for A Kodak DryView Laser Imager. Shop with confidence on eBay!. Kodak DryView Laser Imager with Kodak PACS Link Systems. This product implements all transactions required in the IHE Technical.
|Published (Last):||13 October 2009|
|PDF File Size:||9.59 Mb|
|ePub File Size:||19.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
Use settings through for negative contrast images. Vendor Network Dealer Network.
Kodak, Dryview 8700, Laser Imager
Moving Stored Dryviea — Keypad Press Next, as required, to select Contrast. D Keypad – – Format Menu The numbered descriptions correspond to the numbered callouts in Figure Eastman Kodak Company reserves the right to change this information without notice and makes no warranty, express or implied, with respect to this information.
Use a tack cloth to wipe the surface of the platen, as well as the inside surfaces close to the bottom of the exposure module.
Image size can vary from row to row. Close the cartridge manually refer to page Unprecedented speed and image quality – Up to films per drybiew for maximum productivity – micron laser spot spacing, dpi resolution – Continuous-tone images with up to 4, gray levels – Consistent resolution from application to application – 1: DV Laser Imager.
Removing Jams from Area 3 Weitere Infos zur Lieferzeit in der Artikelbeschreibung. Display Brightness Set Buttons — Sets the keypad display brightness. After erasing all of the stored images, the keypad automatically exits Erase Image mode.
Kodak DryView 8700 Laser Imager Xray Film Processor Imaging System
The center box on the User Settings Menu displays a menu. P Cartridge could not be closed. D Keypad – – Keypad Menu The dryvisw descriptions below correspond to the numbered callouts in Figure Removing Jams from Area 1 Certificates of Compliance and Declarations of Conformity have been issued as shown below.
Laser radiation may be present when the machine operates without panels or covers installed. Instant daylight-load film cartridges. Status messages require no operator action. Examine the 15 images on the contrast test sheet and select the image with the most appropriate contrast.
The functions of the buttons vary depending onthe type of system.
KODAK DRYVIEW LASER IMAGER RÖNTGENFILMENTWICKLER NEUPREIS EURO | eBay
From the console, adjust the window and level controls to achieve the desired image quality on the display monitor. Verifies printed density of each film.
Bis zuletzt in Radiologie im Einsatz. Image Processing Button — Selects smooth or sharp image processing.
K Insert a different film cartridge and press the calibration button on the local panel. This allows the red LEDs to be differentiated from the constant green LEDs which indicate available storage locations. To install the film cartridge, set the leading edge of the cartridge on the cartridge guides, then slide the cartridge into the imager to engage the detents in the bottom of the cartridge.
This dust can create artifacts on printed images. Ensures contrast and density meet preset user preferences.
Imation DryView Dry Laser – Digital Radiographics
The circled numbers in Figure correspond to the numbered steps below. The imager prints the acquired image in a Attention Light — Indicates a condition that may affect imager operation e.
Avoid using a telephone other than a cordless type during an electrical storm. K58 Select a format with fewer images. The accessory should provide fully automatic stitching software that delivers images up to 17 inches wide by 51 inches long 43 x cm.
From the Main Menu, press Imager Menu. D Laser Imager 1 Status: Type B Applied Part! Plug in dryvkew power cord and power up the imager. The film transport drives the film up into the film developer.