programming. HP B Audio The HP B Audio Analyzer and HP E Distortion Analyzer make your audio measurements fast and easy. .. (Does not include HP-IB programming time.) . calibration manual [) and. PDF · HP xA 5xA 7xA 8xA HP-IB Series Programming · HP xA 5xA 7xA 8xA Progamming · HP xA, xA, xA, xA, and. Jun 8, programming instructions when properly installed on that instrument. The HP- IB address switches are located within the Modulation.
|Published (Last):||18 October 2009|
|PDF File Size:||2.10 Mb|
|ePub File Size:||6.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
In addition, rapid-source tuning and rapidfrequency count capabilities not available from the front panel are provided in remote operation. For more information on the Audio Analyzer capabilities, refer to the description in Section 1; TableSpecifications; and TableSupplemental Information. Performance Test 6 Audio Filters. The SPCL key light should remain lit. If any of the recommended equipment is unavailable, instruments with equivalent minimum specifications may be substituted.
This is fast enough for applications such as unsquelching pagers see Figure The 30 kHz key light should light. Seal the shipping container securely. The right display should show between It should be noted that the counter is a reciprocal type; it measures the period of the signal as you do with programmig oscilloscope and computes the reciprocal to obtain the frequency.
B Operation and Calibration Manual | Keysight (formerly Agilent’s Electronic Measurement)
Fast blow fuses may be ordered under HP part numbers1. I then ran the auto detect feature in the NI software to detect the interface and the HP The respective key light should light. Default Conditions and Power-up Sequence. Status Byte Message Description: The right display should show 0. Apart from attentive listening to a hi-fi system, the most intuitive way of analyzing an electrical signal in the audio range is visually with an oscilloscope.
The type of interface card you choose is critical. As shown below, the output data is usually formatted as a real constant in exponential form: For distortion measurements, the Audio Analyzer automatically tunes to and levels the input signal. After nine unsuccessful tries, Error 31 is displayed. Hewlett-Packard further certifies that its calibration measurements progrramming traceable to the United States National Bureau of Standards, to the extent allowed by the Bureau’s calibration facility, prrogramming to the calibration facilities of other International Standards Organization members.
Other than that this program functions similarly to the Power program above. With the exception of pgogramming Rapid Source mode, each code is typically equivalent to a front-panel keystroke in local mode. The left display should show between 26 and 34 kHz. The switch following the Oscillator is normally closed except in the signal-to-noise ratio measurement mode or when an amplitude of OV is entered from the Keyboard.
At this point, one of the two internal plug-in filters can be inserted into the signal path.
The frequency of this signal is also measured by the Counter in the ac level and signal-to-noise ratio measurement modes because of the increased sensitivity at this point. The Floating Output Amplifier converts the single-ended input into a floating signal either output can be grounded or floated up to 1OV peak. Receiving the Data Message. The right display should show a voltage rising from approximately 0 to 1OV in uniform steps.
Index of /~kurt/manuals/manuals/HP Agilent
This is important if the Audio Analyzer is operated near a transmitter or in the presence of large RF signals. Energy available at manypoints may, if contacted, result i n personal injury. This would leave the Audio Analyzer in an unknown state.
Hewlett-Packard and Agiient branded products with the same model number are interchangeable. Thereafter, the invalid code entry is ignored, and subsequent valid entries are processed in normal fashion.
Servicing instructions are for use by servicetrained personnel only.
The local lockout mode prevents loss of data or system control due to someone accidentally pressing maunal keys. Receiving the Pass Control Message. Device listen address received once while REN is true.
The instrument can send the Require Service message in either local or remote mode. Check for compatibility of functions selected 22 Invalid Special Function prefix. Initial Setup The test setup is the same for all of the checks. Source The source covers the frequency range from 20 Hz to kHz.
8903B Operation and Calibration Manual
FLOAT switch provides floating output when required. The Audio Analyzer has dc level as one of its measurement modes.
This error causes a Require Service message to be sent on the HP Thus, trigger codes should always appear at the end of a program string, and the triggered measurement results must be read before any additional program codes are sent.
Time of Return First Measurement: